Before installation, please check the appearance of the cables for damage, and disperse the cables.
1,When using the submersible sensor, avoid the sensor hitting the wall
or other facilities caused by the water flow. If the water flow rate is too fast, the sensor must be fixed.
2 , There should be no obstructions in front of the measurement. For example, fluorescence sensors should be avoided from light.
3,The water depth at which the sensor is installed should not exceed 2 meters from the horizontal, and the water depth at the installation location should not be less than 30 cm according to the fluctuation of the water flow. (Adjustable according to the measured water body)
4, The sensor should be installed in a location where there is no air bubbles to prevent air bubbles from affecting the measurement data of the sensor..
5, Do not use the sensor's own cable to hoist the sensor, and install a cable protection device to prevent excessive tension between the cable and the sensor, causing the cable to break.
6, The sensor should not be installed with the measuring side facing upwards, it can be installed vertically downwards or horizontally.
7, If you use this product in a flow cell, please consult the company's sales staff to confirm the specific installation requirements.